Differential pressure measurements are commonly applied in industrial conditions, due to the fact that measurements that apply them are simple and offer relatively high accuracy. In gas installations, the liquid is often condensed in the form of a droplet present in the gas. When the liquid is transported along with gas, this leads to a significant increase in the differential pressure and incorrect indications of measuring equipment. In addition, the presence of the liquid phase in the flow leads to interference and pressure pulsations. This article reports the results of a study concerned with finding a solution that can offer a way to correct the over-reading of the measured gas flow rate depending on the mass fraction of the liquid in it. The standard orifice was subjected to an experimental study, and then on the basis of this analysis, an algorithm for a computer over-reading model was developed. The experiment involved the measurement of airflow with a small amount of dispersed water in the form of droplets. The results were compared with other correction methods familiar from the literature.
air–water flow, experimental research, over-reading, standard orifice, two-phase flow, wet gas
 Abou El-Azm Aly, A., Chong, A., Nicolleau, F. & Beck, S., Experimental study ofthe pressure drop after fractal-shaped orifices in turbulent pipe flows. ExperimentalThermal and Fluid Science, 34, pp. 104–111, 2010. https://doi.org/10.1016/j.expthermflusci.2009.09.008Figure 6: Results of correction of airflow rate by the model proposed by these authors.
 Annamalai, G., Pirouzpanah, S., Gudigopuram, S.R. & Morrison, G.L., Characterizationof flow homogeneity downstream of a slotted orifice plate in a two – phase flowusing electrical resistance tomography. Flow Measurement and Instrumentation, 50,pp. 209–215, 2016. https://doi.org/10.1016/j.flowmeasinst.2016.07.003
 Bai, B. & Zheng, X., Comparison of the two-phase mass flow coefficient and theover-reading for wet gas flow metering with DP meters. Experimental Thermal and FluidScience, 90, pp. 14–21, 2018. https://doi.org/10.1016/j.expthermflusci.2017.08.027
 Bayazit, Y., Sparrow, E.M. & Joseph, D.D., Perforated plates for fluid management:Plate Geometry effects and flow regimes. International Journal of Thermal Sciences,85, pp. 104–111, 2014. https://doi.org/10.1016/j.ijthermalsci.2014.06.002
 Bertani, C., De Salve, M., Malandrone, M., Monni, G. & Panella, B., State-of-art andselection of techniques in multiphase flow measurement. Report RdS, 67, Torino, 2010.
 Chisholm, D., Flow of incompressible two-phase mixtures through sharp-edged orifices.Journal of Mechanical Engineering Science, 9, pp. 72–78, 1967. https://doi.org/10.1243/jmes_jour_1967_009_011_02
 Chisholm, D., Research note: Two-phase flow through sharp-edged orifices. Journalof Mechanical Engineering Science, 19, pp. 128–130, 1977. https://doi.org/10.1243/jmes_jour_1977_019_027_02
 De Leeuw, R., Liquid Correction of Venturi Meter Readings in Wet Gas Flow. NorthSea Flow Measurement Workshop, Norway, 1997.
 Falcone, G., Hewitt, G. & Alimonti, C., Multiphase flow metering: Principles andapplications.Developments in Petroleum Science, 54, 2009. https://doi.org/10.2118/74689-ms
 Fang, L., Zhang, T. & Jin, N., A comparison of correlations used for Venturi wet gasmetering in oil and gas industry. Journal of Petroleum Science and Engineering, 57,pp. 247–256, 2007. https://doi.org/10.1016/j.petrol.2006.10.010
 Geng, Y., Zheng, J. & Shi, T., Study on the metering characteristics of a slotted orificefor wet gas flow. Flow Measurement and Instrumentation, 17, pp. 123–128, 2006.https://doi.org/10.1016/j.flowmeasinst.2005.08.004
 Hua, C. & Geng Y., Wet gas technique based on slotted orifice and swirlmeter. FlowMeasurement and Instrumentation, 30, pp. 138–143, 2013. https://doi.org/10.1016/j.flowmeasinst.2013.02.001
 Lin, Z.H., Two-phase flow measurements with sharp-edged orifices. InternationalJournal of Multiphase Flow, 8, pp. 683–693, 1982. https://doi.org/10.1016/0301-9322(82)90071-4
 Morrison, G.L., Hall, K.R., Holste, J.C., Deotte, R.E., Macek, M.L., Ihfe, L.M., SlottedOrifice Flowmeter. AIChE Journal, 40(10), October, 1994. https://doi.org/10.1002/aic.690401015
 Morrison, G.L., Hall, K.R., Holste, J.C., Macek, M.L., Ihfe, L.M., Deotte, R.E. &Terracina,D., Comparison of orifice and slotted plate flowmeters. Flow MeasurementandInstrumentation, 5(2), pp. 71–77, 1994. https://doi.org/10.1016/0955-5986(94)90039-6
 Morrison, G.L., Terracina, D., Brewer, C. & Hall, K.R., Response of a slotted orificeflow meter to an air/water mixture. Flow Measurement and Instrumentation, 12,pp. 175–180, 2001. https://doi.org/10.1016/s0955-5986(01)00018-8
 Murdock. J.W., Two-phase flow measurements with orifices. Journal of BasicEngineering,84, pp. 419–433, 1962. https://doi.org/10.1115/1.3658659
 Reodikar, S.A., Meena, H.C., Vinze, R. & Prabhu, S.V., Influence of the orifice shape onthe local heat transfer distribution and axis switching by compressible jets impingingonflat Surface. International Journal of Thermal Sciences, 104, pp. 208–224, 2016. https://doi.org/10.1016/j.ijthermalsci.2016.01.013
 Smith, R.V. & Leang, J.T., Evaluations of correlations for two-phase flowmeters: Threecurrent-one new. Journal of Engineering for Power, 97, pp. 589–593, 1975. https://doi.org/10.1115/1.3446072
 Steven, R., Hall, A., Orifice plate meter wet gas flow performance. Flow MeasurementandInstrumentation, 20, pp. 141–151, 2009. https://doi.org/10.1016/j.flowmeasinst.2009.07.001
 Steven, R., Wet gas metering with a horizontally mounted Venturi meter. FlowMeasurementand Instrumentation, 12, pp. 361–372, 2002. https://doi.org/10.1016/s0955-5986(02)00003-1
 Yadigaroglu, G. & Hewitt, G.F., Introduction to Multiphase Flow Basic Concepts,Applications and Modelling, 2018.
 Zheng, X., He, D., Yu, Z. & Bai, B., Error analysis of gas and liquid flow rates meteringmethod based on differential pressure in wet gas. Experimental Thermal and FluidScience,79, pp. 245–253, 2016. https://doi.org/10.1016/j.expthermflusci.2016.07.017
 Zheng, G.B., De Jin, N., Jia, X.H., Lv, P.J. & Liu, L.B., Gas–liquid two phase flowmeasurement method based on combination instrument of turbine flowmeter and conductancesensor. International Journal of Multiphase Flow, 34, pp. 1031–1047, 2008.https://doi.org/10.1016/j.ijmultiphaseflow.2008.05.002
 Measurement of fluid flow by means of pressure-differential devices- Guidelines forthe specification od orifice plates, nozzles and Venturi tubes beyond the scope of ISO5167, 2007.